arXiv:1610.06682v2 [physics.ins-det] 13 Jan 2017

KATANA - a charge-sensitive triggering system for the SπRIT experiment

P. Lasko^{a,b}, M. Adamczyk^b, J. Brzychczyk^b, P. Hirnyk, J. Łukasik^a, P. Pawłowski^{a,1}, K. Pelczar^b, A. Snoch^c, A. Sochocka^b, Z. Sosin^{b,2}, J. Barney^{d,e}, G. Cerizza^e, J. Estee^{d,e}, T. Isobe^f, G. Jhang^h, M. Kaneko^g, M. Kurata-Nishimura^f, W.G. Lynch^{d,e}, T. Murakami^g, C. Santamaria^e, M.B. Tsang^{d,e}, Y. Zhang^i

aInstitute of Nuclear Physics, Polish Academy of Sciences, Kraków, Poland bFaculty of Physics, Astronomy and Applied Computer Science, Jagiellonian University, Kraków, Poland cUniversity of Wroclaw, Wrocław, Poland dDepartment of Physics and Astronomy, Michigan State University, East Lansing, USA eNational Superconducting Cyclotron Laboratory, Michigan State University, East Lansing, USA fRIKEN Nishina Center, Wako, Saitama, Japan gDepartment of Physics, Kyoto University, Kita-shirakawa, Kyoto, Japan hDepartment of Physics, Korea University, Seoul, Korea iTsinghua University, Beijing, China

Abstract

KATANA - the Krakow Array for Triggering with Amplitude discrimiNAtion - has been built and used as a trigger and veto detector for the SπRIT TPC at RIKEN. Its construction allows operating in magnetic field and providing fast response for ionizing particles, giving the approximate forward multiplicity and charge information. Depending on this information, trigger and veto signals are generated. The article presents performance of the detector and details of its construction. A simple phenomenological parametrization of the number of emitted scintillation photons in plastic scintillator is proposed. The effect of the light output deterioration in the plastic scintillator due to the in-beam irradiation is discussed.

Keywords: charged particle detection, TPC triggering, plastic scintillator, silicon photomultiplier, in-beam radiation damage, number of scintillation photons.

Introduction

The symmetry term in the Nuclear Equation of State (NEoS) is being recently intensively investigated [1–3]. The experimental efforts are focused on constraining the stiffness of the symmetry energy, especially in the range of higher densities [4, 5]. One of the proposed methods consists in getting the information on momentum distribution of charged pions and isotopicallyresolved light charged particles (LCP), Z ≤ 3, emitted in the central heavy-ion (HI) collisions [6–8]. Realization of this kind of measurements inspired the construction of the Time Projection Chamber (TPC) called SπRIT (the SAMURAI Pion-Reconstruction and Ion-Tracker) [9, 10]. It

1Corresponding author. E-mail: piotr.pawlowski@ifj.edu.pl 2Deceased Preprint submitted to Nuclear Instruments and Methods A May 5, 2018

is designed to operate inside the vacuum chamber of the SAMURAI super-conducting spectrometer [11] at Radioactive Isotope Beam Factory (RIBF) of RIKEN [12].

The experiments with the SπRIT TPC at RIKEN are up to now focused on investigation of the central and semi-central collisions at energies of about 300 MeV per nucleon, where nuclear matter densities can reach up to ∼2ρ_0. The target is located about 3 mm upstream of the entrance field cage window inside the SπRIT TPC, so that the HI beam passes through the chamber. To obtain sufficiently large gas amplification for pions and light charged particles, a high electric field is required inside the chamber. The ionization produced by the HI beam would produce an excessive space charge leading to field distortion and could bring a risk of a damage: the charge produced by gas ionization could exceed the safe limit for the pad planes of the TPC. For this reason, a gating grid wire plane was mounted in front of the pad plane [9, 13]. The grid is normally ’closed’ to keep the detector off until a desired collision occurs. When it happens, the opening of the gating grid can occur quickly (in ∼350 ns). Such operation should be performed whenever an incoming projectile breaks up into much smaller fragments.

To produce a logic signal to open the gating grid when a desired event occurs, the KATANA detector, located downstream of the TPC exit window, was constructed. It is designed to play a double role: to produce a minimum bias or majority trigger, and to provide a veto signal whenever a beam particle or a fragment heavier than Z ≃ 20 has passed through the chamber. To fulfill the requirements, the wall has been constructed of two parts, a Veto and a Trigger array. The KATANA-Veto part consisting of 3 thin (1 mm thick) plastic-scintillator paddles with the middle one centered on the beam, has been designed to produce a veto signal for heavy fragments. The KATANA-Trigger array, consisting of 12 thicker (10 mm thick) paddles, arranged on both sides of the beam, has been designed to produce a forward multiplicity trigger.

As the whole detector is expected to work inside the SAMURAI dipole magnet, the apparatus has to be insensitive to magnetic fields up to about 0.5-1.0 T. For this reason the light produced in the plastic scintillators is read-out by silicon photomultipliers. All the materials used for the construction are non-magnetic.

Though the KATANA detector was designed as a triggering device for the SπRIT TPC, it may be used in another experimental context as well. Its versatile, modular and reconfigurable construction makes it easy to adopt other experimental tasks. Currently it is under preparation to work at Bronowice Cyclotron Center in Kraków [14].

This paper will describe the technical details of the KATANA wall construction and of the accompanying electronics. The performance of the detector will also be discussed. In addition, some concepts concerning the modeling of light propagation in scintillators will be presented.

1. Light propagation in thin plastic scintillators

In order to make a realistic simulation of the light propagation in a plastic scintillator some basic ingredients are needed. One of the most important is the mean number of the scintillation photons produced due to the excitation of molecules along the particle track in the scintillator. The process depends in a non-linear way on a specific energy loss of a given charged particle or fragment and on its charge and mass [15]. So far, both the experimental data and the model descriptions of the number of scintillation photons produced due to the light and heavy ions in plastic scintillators are rather scarce. This results in a lack of a realistic description of this process in GEANT4 [16], which is a basic physics simulation environment. To fill this gap, a simple parametrization of the mean number of the scintillation photons produced in a plastic scintillator by charged fragments as a function of their energy, charge and mass has been devised. The details of the parametrization are presented in the Appendix.

The parametrization has been implemented into the GEANT4 to generate the mean numbers of scintillation photons emitted in the VETO paddle during the passage of charged particles. The distribution of charged particles has been generated using the UrQMD transport code [17], with a clustering procedure, for the^{132}Sn+^{124}Sn reaction at E_{LAB} = 300 AMeV. The generated numbers of emitted photons as a function of particle type and its kinetic energy have been used in a Monte Carlo code simulating the light propagation in a scintillator. The real numbers of emitted photons are too large to be followed in a Monte Carlo simulation. For this reason the numbers of scintillation photons served here only to rescale a fixed number of simulated test photons. This procedure allowed to limit the calculation time and to obtain the realistic distributions at the end of the simulation.

In the simulation we considered the propagation of the light generated in a plastic scintillator of a size of 400×100×1 mm^3. For each charged particle hitting the VETO paddle a point-size light source inside the scintillator was assumed. The test photons were supposed to be emitted randomly from this point with an isotropic distribution. Each simulated test photon has been characterized by a weight, an emission time and a wavelength. The weight was here a measure of probability of photon absorption. It was initialized to one at the moment of emission, and reduced during the propagation, depending on the attenuation length and the reflectivity of the walls. The test photon wavelength and the emission time were attributed randomly, according to the characteristics of the scintillator [18].

The light was read out by a pixelated light sensor. Each simulated test photon hitting the sensor area had a certain weight W < 1. As the number of simulated photons per particle was fixed (independently of the particle energy), the weight was scaled by a factor corresponding to the real number of emitted photons. Then the actual number of photons hitting the sensor area was drawn randomly from a Poisson distribution determined by the weight. The response of the light sensor is governed by the photon detection efficiency (PDE) function related to the wavelength of the photon [19]. If the photon is accepted according to the PDE function, a randomly chosen pixel is activated. If the pixel had not been activated before, it is marked as active, and produces an electric signal of 5 ns fall time. The sum of all signals coming from all light sources and all light sensors in a paddle was then digitized with a software replica of a 14-bit, 500 MS/s Flash ADC. The amplitude of the digitized pulse was considered as a final output of the detector.

In the simulations a variety of light readout methods were considered. The best found solution is schematically presented in Fig. 1 (see Sec. 2 for more information on technical details). The light is first collected along two shorter edges of the plastic bar by a wavelength shifting fiber (WLS) [20]. The WLS fiber is a special plastic absorbing light and re-emitting it isotropically in another spectral length. On both ends of the fiber the light sensors are mounted. This solution was found to be less sensitive to the hitting-particle position than in the case of light sensors coupled directly to the edge of the scintillator.

The main issue studied in the simulation was the efficiency of charge filtering basing on the light signals generated by heavy fragments in the VETO paddles. Figure 2 shows the predicted performance of the trigger/veto system. The events generated by the UrQMD model were filtered by the condition that at least one charged particle hits one of the 12 thick (triggering) paddles and the VETO paddles do not produce the signal higher than the value corresponding to Z ∼ 20. The figure presents the efficiency of this triggering system as a function of the charge of the particle hitting the VETO paddles (a), and as a function of the impact parameter (b). 3

Figure 1: Schematic view of a VETO paddle

Figure 2: Simulated trigger efficiency of the KATANA detector: a) acceptance for the charged fragments hitting the VETO paddles; b) distribution of the accepted impact parameters.

The upper panel shows a very good performance of veto paddles in cutting-off the heavier fragments using the amplitude threshold. Obviously, the most peripheral collisions are cut-off by the VETO signal. This strong influence on impact parameters accepted by the trigger is displayed in the lower plot. 4

2. Technical design of the detector

A technical design of the KATANA detector is presented in the upper part of Fig. 3 while the bottom part presents its realization. Fifteen scintillators, each 10 cm wide and 40 cm high, are mounted on a rectangle, aluminum frame. Each scintillator is wrapped in a sheet of a lightreflective, metalized Kapton foil. Three veto plastic bars, marked with dashed lines, (size 400 × 100 × 1 mm^3, made of BC-404) are placed in the middle of the frame. Twelve trigger plastics (400 × 100 × 10 mm^3, made of BC-408) are placed unevenly on both sides of the beam (7 on the left and 5 on the right side) to account for bending of the charged particles in the magnetic field.

Figure 3: Top panel: a technical drawing of the KATANA detector with the external dimensions in mm. Bottom panel: a photo of the detector before mounting it inside the SAMURAI vacuum chamber.

Each thick trigger paddle is fixed to the frame by two handles. The scintillation light is collected by two light sensors located in the middle of the top and bottom edge of the plastic, and is read out directly without any light guides. The thin veto paddles have more complex design (see Fig. 4). In this case the light is read out by four light sensors. First, the light is collected along the plastic edge with the use of a WLS fiber (square cross section of 1 × 1 mm^2, type BCF-92). The WLS collects light from shorter sides of the scintillator and propagates it to the light sensors mounted on both sides of the fiber (see Fig. 1). The fibers and the light sensors were mounted applying an optical grease as an interface. The main physical parameters of the plastics are presented in Tab. 1.

In both cases (thick and thin paddles), the Multi Pixel Photon Counter (MPPC) devices from Hamamatsu [19] have been used as the light sensors. It is worth mentioning that each MPPC is 5

Figure 4: Veto scintillator handle.

Table 1: Main physical properties of the plastic scintillators (BC-408 and BC-404) [18] and WLS light guide (BCF-92) [20].

BC-408 BC-404 BCF-92 Emission maximum, λ_{max} (nm) 425 408 492 Attenuation length, L (cm) 210 140 >350 Refractive index, η 1.58 1.58 1.42 Light velocity, v (cm/ns) 18.9 18.9 21.1

mounted directly on a preamplifier PCB to minimize the noise level. The front-end electronics is completed with analog adders providing the analog sum of signals coming from all sensors of a single paddle. Thus, each paddle produces finally a single electric pulse. An inverted copy of the signal is available as well.

The frame of the KATANA detector is mounted on two trolleys, capable to move on special rails used primarily to set-up the position of the SπRIT TPC inside the SAMURAI vacuum chamber. The trolleys allow also for left-right adjustment of the frame.

3. Electronics

3.1. Light sensors

As mentioned above, the light readout has been done using two types of MPPC. For the thick paddles the S12572-025P product was used. Its photo-sensitive area of size 3 × 3 mm^2 consisted of 14400 parallelized pixel avalanche diodes. The thin paddles were read out by the S12571-010P element containing 10000 pixels on 1 × 1 mm^2 area. Main parameters of the applied MPPCs are specified in Tab. 2.

3.2. Preamplifier

The circuit of the preamplifier applied for MPPCs is shown in Fig. 5. It consists of a common base and a common collector (emitter follower) amplifiers based on the T1 and T2 transistors, respectively, and a simple pole zero cancellation circuit based on the C3, R7 and R8 elements.

Table 2: Main parameters of the applied MPPCs [19].

S12571-010P S12572-025P Photosensitive area (mm^2) 1×1 3×3 Number of pixels 10000 14400 Typical operating voltage (V) 69 68 Gain 1.35 × 10^5 5.15 × 10^5 Peak sensitivity wavelength (nm) 470 450 Gain temp. coefficient (K^{-1}) 1.6 × 10^3 8.2 × 10^3 Peak photon detection efficiency (%) 10 35

The low voltage power supplies are omitted in the schematic but it is obvious that a high stability and low noise power supply is required.

Figure 5: Schematic of the preamplifier for MPPC.

The MPPC (D1) is polarized by sufficiently high voltage (HV) provided by HV power supply described below. High frequency signal pulses are passing through C1 to the low impedance input of the common base amplifier. This kind of amplifier was chosen because of its good characteristics for high frequency operation.

A common collector amplifier was used to match the output impedance. The R6 and R9 resistors also play a significant role in this respect. The C2 capacitor cuts off the DC component.

The preamplifier allowed to achieve a one-photon resolution in amplitude. Figure 6 shows the amplitude distribution obtained from a thick paddle stored in a dark room. The peaks correspond to events with one, two and more photons. It clearly shows sensitivity and amplitude resolution of the thick (trigger) paddles. Of course, during the experiment, the thresholds were set-up much above this natural noise to detect only light charged particles. On the other hand, the sensitivity of thin (veto) paddles is much lower, as they should give response only for heavy charged fragments.

The preamplifiers were optimized to obtain a gain sufficient to observe signals from detectors in the range 0 − 2 V. Because of different optical characteristics of the small (1 × 1 mm^2) and big (3 × 3 mm^2) MPPCs used for thick and thin paddles, the preamplifiers were prepared in two variants. Table 3 presents the parameters of the applied elements.

Table 3: The elements of the preamps applied for small and big MPPCs. See Fig. 5 for the meaning of the symbols.

Element 1 × 1 mm^2 MPPC 3 × 3 mm^2 MPPC D1 S12571-010P S12572-025P R1 1kΩ 1kΩ R2 60.4Ω 1kΩ R3 56Ω 50Ω R4 750Ω 1.5kΩ R5 750Ω 5kΩ R6 10 kΩ 2 kΩ R7 51Ω 2kΩ R8 4.7Ω 0 R9 - - R10 110Ω 0 C1 100nF, 100V 100nF, 100V C2 10nF 10nF C3 100nF 100n F C4 100pF - C5, C6 1μF, 10V 1μF, 10V T1, T2 BFR106 BFR106

3.3. High voltage power supply

The MPPC elements need to be biased. This goal was achieved by designing a dedicated HV power supply, satisfying the following special requirements: 10 mV precision, low noise and low voltage ripple (< 1 mV), 60 − 80 V range of regulation, 40 channels, remote control via Ethernet, multi-point temperature control, current limiter and a feedback information (saturation/error). The device is controlled by a STM32F103 micro-controller with ARM Cortex-M3 core. A separate Ethernet controller allows to create a system working in a server mode. The IP and port numbers for the communication are configurable. The interface for the DAC cards and HV cards was custom designed and allowed to connect those cards in parallel. Each card is provided with switches for setting the card address.

The HV supply is temperature sensitive. In order to monitor the temperature a DS18B20 element has been chosen as a built-in temperature sensor. This is a digital thermometer with 12-bit (0.0625^∘C) resolution and 1-Wire interface for communication. Two such devices are installed inside the high voltage power supply box. Additionally, six such sensors are mounted on the frame of the KATANA detector to monitor the operating temperature of the MPPCs. The temperature data are collected by the controller and are available upon request from the server.

3.4. Signal processing

The amplified MPPC signals of each plastic scintillator are summed, providing one analog signal for each scintillator. Each analog adder has normal and inverted output. In total KATANA provides 15 normal (positive) signals and 15 inverted (negative) ones. The positive signals are sent to a Flash ADC board (CAEN V1730) for control purpose. The negative signals are dispatched to the leading edge discriminators and are compared to the threshold levels set by remotely controlled DAC cards. The logic outputs of discriminators are then analyzed by a logic 8

Figure 6: Single-photon amplitude resolution achieved for thick paddles equipped with S12572-025P elements.

circuit built on a FPGA board, providing majority trigger for the 12 triggering KATANA paddles (with the majority threshold adjustable between 1 and 12). The FPGA circuit takes also into account the other logic signals used in triggering. Several Gate&Delay units are programmed in FPGA to synchronize the signals coming from different sources before performing logic operations (coincidence/anti-coincidence). KATANA multiplicity threshold, delay durations and gate widths can be remotely controlled via a RaspberryPi controller [21].

20-channel discriminator board, FPGA board, logic output buffers and RaspberryPi are integrated in a remotely controlled “Trigger Box”. An integrated structure of the Trigger Box together with the fast and flexible FPGA logic allows to provide fast signals (within < 100 ns) required by the TPC gating grid and by the acquisition system.

4. Detector efficiency

The efficiency for HI detection and fragment discrimination is a crucial characteristic of the KATANA-Veto paddles and has been tested using the 300 AMeV^{132}Xe beams at HIMAC (the Heavy Ion Medical Accelerator in Chiba). The test setup is depicted in Fig. 7.

Figure 7: Test run setup at HIMAC.

The prototype of the KATANA-Veto module was placed directly in the beam, in air. A KRATTA detector module [22] was mounted behind the veto paddle. The KRATTA module is a triple telescope allowing to identify the charge of a hitting particle. For the test, the gain 9 of the KRATTA module was lowered to increase its charge identification range. Using it as a reference takes advantage of its high detection efficiency and its charge resolution. The charge spectra collected by the KATANA-Veto prototype and the KRATTA module in coincidence are shown in Fig. 8. In both cases the charge resolution for the elastically scattered beam particles could be estimated. The FWHM of the beam peaks amounts to about 0.6 and 1.3 charge units in the KRATTA and the KATANA case, respectively. The former reflects the resolution of a 1 mm thick silicon detector and the latter the resolution of 1 mm thick BC-404 plastic scintillator. In the case of the KRATTA detector the observed peaks correspond to resolution of single charges.

Figure 8: Charge resolution obtained at HIMAC with the KATANA prototype plastic bar (hatched histogram) and with the KRATTA module (black line) for the^{132}Xe fragmentation at 300 AMeV. Note the single charge resolution for the KRATTA detector.

The efficiency of the veto module for the HI discrimination could be determined by correlating the signal amplitude from the veto paddle with the fragment charge number provided by the KRATTA module. Figure 9 shows the detection efficiency of the VETO paddle as a function of the hitting fragment charge for various threshold settings on the discriminator. The fragment charge selectivity of the KATANA veto paddle is clearly visible. We can also conclude that the simulated result presented in Fig. 2a is realistic.

5. Radiation damage

As the veto paddles are exposed to the beam directly, a significant drop of their efficiency of conversion of the absorbed energy into light may be expected due to the radiation damage. During the SπRIT experiment a cocktail of beams around^{108}Sn and^{112}Sn isotopes with an average intensity of ∼7.0 kHz was applied. Even for such a low beam rate a decrease of light output is expected after dozens of hours of exposure. The damage is reflected in the decrease of the amplitude measured for the pure beam events. Figure 10 presents this effect as a function of the experimental time. The amplitude decrease is about 1% per beam day. On the other hand, no significant effect on the width of the beam peak was observed.

It should be underlined that the data used for the radiation damage studies do not come from a dedicated and optimized experiment. Rather, they are monitoring data taken with the use of the KATANA triggering system during the main SπRIT experiment. This may be the reason for a large scatter of the data points corresponding to the individual runs with possible changes 10

Figure 9: Measured efficiency of a thin VETO paddle. Different symbols (squares, points, triangles and stars) correspond to increasing thresholds applied to the VETO signal amplitude.

Figure 10: In-beam aging of a 1mm thick BC-404 plastic scintillator. The points show the mean amplitudes obtained for the beam particles (^{108}Sn and^{112} Sn at 300 AMeV). The line shows a linear fit to the points. The histograms show the local temperature variation at the top (upper histogram) and at the bottom (lower one) of the middle Veto paddle.

in the beam conditions (position, composition, intensity). The two histograms at the bottom of Fig. 10 show the variation of the local temperature measured at the top (upper histogram) and at the bottom (lower one) of the middle Veto bar. The temperature at the top was about 0.6^∘C higher because of the proximity of the TPC electronics. The sharp spikes around the middle of the beam time indicate the presence of the experimenters in the vicinity of the sensors during the beam breaks. The gain-temperature coefficient for the applied sensor (see Tab. 2) implies the gain change of about 1%/^∘C. Thus, the observed relative stability of the temperature near the MPPC sensors excludes the temperature effect on the MPPC gain that might potentially contribute to the observed amplitude variation.

Summary

In summary, we have presented technical details and performance of the KATANA array. The detector was designed to provide trigger/veto signal for the SπRIT experiment at RIKEN. It was playing a significant role in the detection system, protecting the TPC against a risk of damage by excessive ionization produced by HI beams. Fast response, high veto and trigger efficiency, insensitivity to magnetic fields, stability, portability and the possibility of remote control are the main attributes of this array.

Acknowledgments

This work was supported by the Polish National Science Center (NCN), under contract Nos. UMO-2013/09/B/ST2/04064 and UMO-2013/10/M/ST2/00624 as a part of the SπRIT Project supported by the U.S. Department of Energy under Grant Nos. de-sc0004835, de-sc0014530, de-na0002923, US National Science Foundation Grant No. PHY-1565546 and the Japanese MEXT KAKENHI (Grant-in-Aid for Scientific Research on Innovative Areas) grant No. 24105004. The authors are indebted to HIMAC and RIBF technical staffs for an excellent beam time.

Appendix: Mean number of scintillation photons

In order to obtain a realistic number of scintillation photons hitting the light sensor in a simulation one needs first of all a realistic number of photons produced along a particle track. The realistic number of photons hitting the light sensor is particularly important when simulating the response of pixelated light sensors. This is because the simulation has to provide an optimal number of pixels assuring the required dynamic range of the device, or in other words, the number of pixels sufficient for detection of the maximum ionizing fragment before the sensor saturates. A fixed number of scintillation photons per MeV, independent of the energy and of the ion identity, offered by GEANT4 seems to be an oversimplification and was the main motivation to devise a more realistic energy–light conversion formula.

The experimental data on the number of scintillation photons produced in plastic scintillators are rather scarce. The most comprehensive set of data on a relative light output for various ions at intermediate energies is presented in [23] and refers to the NE-102 scintillator. The authors provide simple parameterizations of the light output as a function of the incident energy and the mass and atomic numbers, however these are very approximate. Attempts to better describe the above data are presented in [24, 25]. There, the models are able to better describe the data, however the differential form of the obtained parameterizations is not best suited for Monte-Carlo simulations and energy calibrations. Instead, we attempted to obtain a simple parametrization of the light output assuming the following formula:

N(E, A, Z) = a_0 � E^{a4} − a_1A^{a6} ln(1 + E^{a4} ) − a_2A^{a6} Z^{a7} ln � 1 + E^{a4} a_3A^{a6} Z^{a7}

�_{a5} (1) where N is the mean number of scintillation photons produced in a scintillator and E, A and Z are the energy in MeV, and the mass and atomic numbers of an incident ion, respectively.

The following parameter values have been obtained by fitting the formula (1) to the relative light output data for the NE-102 scintillator: a_0 = 89.435(×219), a_1 = 0.581, a_2 = 0.0914, a_3 = 0.29, a_4 = 0.34, a_5 = 3, a_6 = 0.193 and a_7 = 2.762. An additional factor of 219 has to be 12

Figure 11: The mean number of scintillation photons produced in a plastic scintillator by charged particles of a given incident energy. The squares represent the absolute numbers of photons emitted due to protons incident on a BC-400 plastic scintillator [26]. The open and filled circles represent the experimental data of [23] for^1H,^2H,^3He,^4He,^6Li, 7Li,^{12}C,^{16}O,^{32}S,^{40}Ca and^{81}Br from top to bottom, respectively. The original relative light output for these data points has been multiplied by a factor of 219 to match the absolute numbers of photons represented by the squares. The solid lines represent the fits of the formula (1) to the data. The dashed lines connect the data points for a given incident ion.

applied to the overall scale factor, a_0, in order to match the proton data for the NE-102 with the absolute numbers of scintillation photons obtained for protons in the BC-400 scintillator [26].

The form of Eq. (1) has been inspired by the results of [27] which, apart from the δ-electron term, have been obtained for the Birks’ formula assuming the 0-th order Bethe-Bloch approximation: ΔE ∝ A Z^2/E. Its extension by introducing an additional logarithmic term and applying exponents to the energy, number of photons, mass and atomic numbers is purely phenomenological. The quality of the fit is presented in Fig. 11, together with the scaled experimental data of [23] and with the data for protons from [26]. The χ^2 of the fit amounts to about 0.8. The main motivation for devising the above formula was to enable realistic simulations of the number of scintillation photons produced in plastic scintillators, however its functional form (or its inverse) may possibly also be used as an energy–light conversion formula for the energy calibration of plastic detectors.

References

[1] M. B. Tsang et al., Phys. Rev. C 86, 015803 (2012). [2] B.-A. Li, and X. Han, Phys. Lett. B 727, 276 (2013). [3] C.J. Horowitz et al., Jour. of Phys. G 41, 093001 (2014). [4] P. Russotto et al., Phys. Lett. B 697, 471 (2011). [5] P. Russotto et al., Phys. Rev. C 94, 034608 (2016). [6] B.-A. Li, Phys. Rev. Lett. 88, 192701 (2002). [7] J. Rizzo et al., Phys. Rev. C 72, 064609 (2005). [8] J. Hong and P. Danielewicz, Phys. Rev. C 90, 024605 (2014). [9] R. Shane et al., Nucl. Inst. Meth. A 784, 513 (2015). [10] G. Jhang et al., J. Korean Phys. Soc. 69, 144 (2016). [11] T. Kobayashi et al., Nucl. Inst. Meth. B 317, 294 (2013). [12] Y. Yano, Nucl. Inst. Meth. B 261, 1009 (2007). [13] S. Tangwancharoen et al., arXiv:1612.06708. [14] https://ccb.ifj.edu.pl/en.home.html 13

[15] J.B. Birks, Proc. Phys. Soc. A 64, 874 (1951); Phys. Rev. 86, 569 (1952). [16] S. Agostinelli et al., Nucl. Inst. Meth. A 506, 250 (2003). [17] H. Petersen et al., Phys. Rev. C 78, 044901 (2008).

[18] BC-400,BC-404,BC-408,BC-412,BC-416 Premium Plastic Scintillators, Data Sheets, Saint-Gobain. [19] http://www.hamamatsu.com/jp/en/product/category/3100/4004/4113/index.html [20] Scintillating Optical Fibers, Data Sheets, Saint-Gobain. [21] http://www.raspberrypi.org [22] J. Łukasik et al., Nucl. Instr. Meth. A 709, 120 (2013). [23] F.D. Becchetti et al., Nucl. Inst. Meth. 138, 93 (1976). [24] N. Matsufuji et al., Nucl. Inst. Meth. A 437, 346 (1999). [25] A. Menchaca-Rocha, Nucl. Inst. Meth. A 602, 421 (2009).

[26] http://www.crystals.saint-gobain.com/uploadedFiles/SG-Crystals/Documents/SGCOrganicsBrochure.pdf [27] M. Pârlog et al., Nucl. Inst. Meth. A 482, 693 (2002).